This package comes with a BiXPower BP100 high capacity (96 Wh) lithium ion rechargeable battery pack and a 24V DC converter specially designed for a ResMed AirSense 10 CPAP machine. It can power ResMed AirSense 10 CPAP machines for 1 ~ 2 nights with the humidifier turned off.
The battery pack comes with one BP100 battery + One ResMed S10 DC Converter + One Port I output cable + AC Charger.
To use this battery with ResMed CPAP DC converter, just plug in the ResMed DC converter to Port II with enclosed connector tip like this:
Port I (16/19/24V port) may not supply enough Amps surge current to start a CPAP machine. Please use Output Port II ( 9 ~ 12V Port) to power a CPAP machine . Just connect S10 converter to battery Port II.
Its compact size allows it to easily fit into a laptop/notebook carry bag.
This battery is made with high quality Panasonic Li-ion battery cells and is compatible with many devices that need 15V, 16V, 16.5V, 18V, 18.5V, 19V, 19.5V or 20V DC power. The 24V DC converter is specially designed for the ResMed CPAP Machine as it can convert the battery output voltage to 24V DC power. The 5V USB port can power many small devices such as smart phones, tablets.
|Input Charging Port:||Input Voltage Range: DC 12 ~ 24V. Connector Size: 5.5 x 2.5mm|
|Outputs:||Port I: |
Output Voltage: 16V, 19V or 24V selectable
Output Connector Size: 5.5 x 2.5mm
Maximum Output Current: 4A @16V, 3.5A@19V, 2.5A@24V
Port II: Nominal Output Voltage 11.1V
Output Voltage Range 9V ~ 12.6V
Output Connector Size: 3.5 x 1.35mm;
>Maximum Output Current: 5A
Port III: 5V USB Port; Maximum Output Current: 2.A
Total maximum output with all 3 outputs: 65W
|Charging time /Temperature||4-6 hours / 0～45 ℃ |
|Size||180 x 110 x 22.5mm ( or 7.2" x 4.2" x 0.9")|
|Weight:||621g ( or 1.4 lb)|
|Power Indicator:||4 level LED indicator|
|Built-in Protection Circuit:|
-Over-charge ; Over-discharge; Over current ;Over temperature
- Short circuit protection
-20～ 60 ℃
- 20～40 ℃
This Universal External Lithium Rechargeable Battery Pack can provide power for laptop computers, digital cameras, digital photo frames, PDAs, cell phones, DVD players.
This battery features 3 output ports.
Output Port I
Output Port I features 3 different regulated voltages through a hardware volt selection switch: 16V, 19V, 24V. This port can be used to power laptop notebook computers. Please set the port I output voltage close to your laptop power adapter output voltage or laptop's input voltage.
All 3 voltages will work with ResMed S10 DC Converter, so you may set Port I at any voltage for S10 DC converter.
To change the output voltage, just simply move the volt selection switch.
Output Port II & Port III
Port II's output voltage range is 9V to 12.6V. Nominal voltage is 11.1V. When the battery is fully charged, its voltage is 12.6V. Output voltage will then decrease gradually. Most of the time, the output voltage will be around 10V ~ 11V ( nominal voltage is 11.1V). When the voltage drops to 9V, the port will shut off and stop providing power. This port can be used to power portable DVD players and digital photo frames.
Use Port II for CPAP DC Converter. The S10 converter will convert this port 9 ~ 12V voltage to 24V Voltage needed for CPAP machine.
Output III is a 5V USB port. Its maximum output current is 2.1A, which can be used to power or charge iPad, tablet computers and any other device that can be charged from a USB port (USB power cable not included).
Charge the battery: This battery can be charged from any 100V ~ 240V, 50Hz/60Hz AC power source with the battery AC charger.
This battery has built-in charging controller to control the charging process. It can accept a very wide range of voltage for charging. You can use a regular power adapter with 12V to 24V DC power to charge the battery. The battery charge port is a 5.5 x 2.5mm female jack, therefore a 5.5 x 2.5mm male connector is required.